Posted 11/16/2023 12:56 (#10484420 - in reply to #10483115) Subject: RE: Trimble Field-iq and Muller presure sensor 30303591
www.TME.eu in Lodz, Poland has a good selection of WIKA and other quality pressure transducers.
Most of them are industry standard 8 - 30V input and 4-20mA output. Lots of WIKA sensort around 100 €/$.
I ususally get all the connectors etc from there.
Low cost chinese can be found from ebay and Aliexpress.
To get 1-5V from 4-20mA pressure signal You just simply use a 250 ohm resistor between signal and signal ground connectyors of your Field-IQ.
I would imagine that your Arrag sensor 4-20 mA 20 bar should be just fine.
I will enclose a simple scheme how to wire the sensor and the resistor and also a Field-IQ paper with usefull information of pinouts etc.